Turn the scale to its operating position and remove shipping wires wrapped around the center of the lever system.
Then, reinstall the platform removed in para c, above.
Set the scale in place on scale frame and level. (Refer to Figure 4-4e )
Turn the four adjustable feet (7) until the scale is resting firmly at each corner on the supporting surface.
Verify the scale is level by observing the leveling device attached to the side of the column assembly. The
plum rod should be centered as closely as possible for the best results.
Using the lock nuts loosened in para d, above, lock each adjustable foot in position.
Adjust the zero balance device (3, Figure 4-5) to obtain a zero reading. Turning the zero balance clockwise
causes the indicator assembly to move clockwise. Turing the zero balance device counterclockwise causes the
indicator assembly to move counterclockwise.